Recorded Webinar (April 2019) | Bcl2Fastq is a Linux-based software that converts base call files generated from an Illumina sequencing run to FASTQ files as well as demultiplex samples. This webinar focuses on Bcl2fastq v2+ and is targeted for new to intermediate users who are interested in the configurable options that this command line program allows. We will cover the following topics: requirements to run Bcl2Fastq, command line basics, commonly used options, frequently asked ‘How To….’ questions.
In this video, an Illumina Field Applications Scientist discusses what to expect during the pre-installation, shipping, and delivery stages. We’ll highlight important steps in the process and introduce you to available resources. Note that, depending on your region, the steps shown in this video may not apply to the iSeq 100 system.
Recorded Webinar (July 2019) | Illumina next-generation sequencing technology allows for massive parallel sequencing. Illumina Technical Support invites you to join us for a presentation and discussion on sample preparation, cluster generation and sequencing by synthesis chemistry. This webinar is targeted to those new to next-generation sequencing and will discuss the following topics: Illumina library construction, cluster generation methods by platform, sequencing by synthesis, and multiplexing and primary analysis.
Upcoming Webinar | Illumina next-generation sequencing technology allows for massive parallel sequencing. Illumina Technical Support invites you to join us for a presentation and discussion on sample preparation, cluster generation and sequencing by synthesis chemistry. This webinar is targeted to those new to next-generation sequencing and will discuss the following topics: Illumina library construction, cluster generation methods by platform, sequencing by synthesis, and multiplexing and primary analysis.
Jan 09, 2020 PST
This course provides an introduction to Illumina Experiment Manager, and shows you the steps to create sample plates and sample sheets.
This course discusses the purpose of ExAmp cluster amplification, lists the steps in the ExAmp cluster amplification workflow, and provides best practices for preparing the ExAmp reaction.
Learn about the benefits of instrument performance monitoring through Illumina Proactive, a service in which instrument performance data are sent to Illumina to enable proactive maintenance.
This video discusses how to choose the appropriate sequencing primers for your run. Since commonly used Illumina indexes, TruSeq™ and Nextera™, utilize different index primer site sequences, it's important that the correct primer is loaded into the HiSeq for sequencing.
This video offers tips and troubleshooting advice related to the HiSeq Instrument computer, including navigating the instrument's Port Manager, formatting hard drives, and setting up a second run.
This video offers tips on the mechanical and thermal systems of the HiSeq, as well as best practices on maintaining your instrument.
This video introduces the fluidics system. Topics include fluidics leaks, syringe pump function, and priming the HiSeq 2500 template loading station.
This video continues to share tips and tricks for the fluidics system started in Part 4 of the series. The video focuses on how to identify and remove blockages from the fluidics lines to ensure the HiSeq instrument is used at its highest capacity.
This video shares IT requirements for implementing Illumina Proactive, an instrument performance monitoring service provided by Illumina that can increase instrument uptime, improve operational efficiency, and reduce the risk of lost resources.
An Illumina Field Applications Scientist describes what it means to sequence through the insert of your prepared library. We look at Bioanalyzer traces to assess the range of insert sizes in a library.
This video shows you how to use Sequencing Analysis Viewer (SAV) plots to determine if your sequencing insert is too short. We also explore how to use the adaptertrimming.txt file on the MiSeq to diagnose short inserts.
This video provides an overview of overclustering and how it can impact your sequencing data. An Illumina Field Applications Scientist shows you how to use Sequencing Analysis Viewer (SAV) to look for common symptoms of an overclustered HiSeq or MiSeq flow cell.
Recorded Webinar (September 2019) | llumina Technical Support invites you to learn about BaseSpace, Illumina’s cloud-based sequencing data analysis solution. In this presentation, we will discuss how to navigate the BaseSpace dashboard, general BaseSpace functionally, and tips and tricks for successful application and workflow utilization. We will cover the following topics: How to upload/view data in BaseSpace, discussion of major analysis workflows, general overview of how to analyze data on BaseSpace.
Please note: BaseMount and BaseSpace CLI are alpha software releases; features and functionality may change over time, and support is directly through Illumina software development.
Recorded Webinar (November 2019) | The Sequencing Analysis Viewer (SAV) Software is an application where users can view important quality metrics generated during sequencing runs. This webinar will provide a guided tour for beginners on how to use SAV, as well as tips and tricks for reviewing the most useful information for sequencing runs. This webinar is targeted at new users and will go over the following topics: how to load data into SAV, what metrics are available in each tab of the software, and understanding which are the most valuable metrics for run review and where to find them.
Upcoming Webinar | Low diversity libraries, such as those used for amplicon sequencing, can present challenges on Illumina sequencing systems and may skew software performance and data accuracy resulting in poor quality runs. This webinar is ideal for new and intermediate users interested in considerations for the sequencing of low diversity libraries and will cover the following topics: understanding and visualizing base diversity in Sequencing Analysis Viewer (SAV), strategies to optimize low diversity sequencing data, % PhiX spike-in and cluster density recommendations for different sequencing platforms, and an overview of base calling and quality score calculations.
Jan 23, 2020 PST